Program Routine Maintenance Of The Glove Box
Etelux reminds you that in addition to the frequent inspection of the glove box, the daily maintenance operate of the glovebox is also important
one. Right before and after each operate, very first cleanse the atmosphere to make the ground dry and clear. Then clean the gear, together with the inner and outer surfaces of the glove box, the observation technique, the function floor, and so on., to be free of charge of debris and clean up.
2. Thoroughly document the drinking water, oxygen information, and gas consumption of the machines in the glove box each day to steer clear of insufficient gasoline force and lead to the products to cease running.
three. Pay attention to the quantity of gasoline in the operating fuel cylinder at any time and replace it in time.
4. In scenario of electrical power failure, press the "vacuum pump" button on the panel to make the vacuum pump operate soon after energy on
5. Be aware: Immediately after energy on, if there is a issue that it simply cannot start off, the likelihood is that the vacuum pump ability cable is loosely linked. The next rationale is that the vacuum pump’s personal change is not turned on
6. When you want to put the sample into the Nuclear Glove Box box, use a tiny transition chamber as considerably as doable. Ahead of opening the transition doorway, fork out consideration to irrespective of whether the tension gauge pointer on the valve cavity is at the zero place. If it is, you can open up it specifically if not, you will need to flip the change beneath the valve cavity to the "filling" posture to enable the gasoline enter the glove box changeover compartment, and then flip the swap back again to the intermediate posture, then you can open the transition compartment Set the sample in the doorway, and then shut the door. When inserting samples, be mindful not to contact the internal wall of the transition chamber with your fingers, wear gloves to choose and area the sample
Doing these routine servicing responsibilities can cut down the reduction price of the glove box, increase the provider existence of the glove box, and also reduce the prevalence of accidents